This file is
Feel free to add/update/remove items on this page.
Common to All Fluorinert Chillers:
Temperature triggered Auto-shutoff
Barrel Fluorinert Chiller:
EndCap Fluorinert Chiller:
Needs replacing every few years...will probably last the
life of the experiment.
Spare Fluorinert Chiller:
Pump: Speck Pumpen
60 Hz, 76 Litres/min
Same as EndCap chiller cartridge.
Chiller parameters on the chiller panel are all screwed up! What
First, check to make sure the cables are plugged in correctly. GMB is
in the same crate as emc-mon and labeled. The connector to look for is
a DB-25 (bottom connector). Chiller side is a DB-9 that plugs into a side
of the power box.
Make sure the GMB is functioning correctly. The lights on the GMB
should blink every 10 seconds or so.
Lastly, restart the chiller. This is obviously not a desired
thing to do, so PLEASE make sure you've exhausted all other means.
Spare chiller flow rate fluctuating:
There is nothing wrong with the meter. This happens
when the valve is not open all the way. This chiller
has shown some instabilities in flow rate at the